# Synopsis: Correcting Hardware Bias in Molecular Spectrometers

More accurate measurements of ${\text{CO}}_{2}$ in the atmosphere can be obtained by accounting for discrepancies in spectroscopy hardware.

Infrared spectroscopy is an irreplaceable tool for quantifying atmospheric ${\text{CO}}_{2}$, but obtaining the gas concentration from a spectral signal depends on knowing the absorption cross section—how strongly the molecule interacts with photons at a particular wavelength. Researchers have long sought to determine the ${\text{CO}}_{2}$ cross section for infrared wavelengths with a precision better than 0.1%, but even calculations from first principles come with an uncertainty above 1%. Now, by measuring a reference sample—whose composition was precisely known —with multiple highly sensitive cavity ring-down spectrometers (CRDSs), Adam Fleisher and colleagues at the National Institute of Standards and Technology in Maryland, have derived the ${\text{CO}}_{2}$ cross section with an uncertainty of 0.06%.

In a CRDS, a laser beam bounces between a pair of mirrors and is gradually absorbed by the gas in the cavity, with the rate depending on the gas concentration. After minimizing known sources of error in a set of exquisitely precise CRDSs, Fleisher and colleagues were able to focus on the performance of the digitizer hardware—the part of the instrument that turns the analog output from the photodetector into a digital signal for analysis. When they compared the results of five separate digitizers, the researchers found unexpected biases unique to each device. By adjusting their measurements based on this previously uncharacterized digitizer bias, the team obtained a value for the absorption cross section with a 25-fold improvement in uncertainty relative to earlier results.

The improvement equates to a ${\text{CO}}_{2}$ measurement precision of better than 0.4 parts per million at current atmospheric concentrations, which will allow more accurate characterization of gas sources and sinks. The result also applies to studies of other gas molecules, including the optical detection of biosignatures on exoplanets.

This research is published in Physical Review Letters.

–Marric Stephens

Marric Stephens is a freelance science writer based in Bristol, UK.

More Features »

### Announcements

More Announcements »

## Previous Synopsis

Topological Insulators

## Next Synopsis

Statistical Physics

## Related Articles

Nanophysics

### Synopsis: Imaging Water Molecules on Metal

Atomic force microscopy reveals the structure of a single layer of water molecules adsorbed on a nickel surface, potentially expanding our understanding of catalysis.   Read More »

Atomic and Molecular Physics

### Focus: Cooling on the Negative Side

A new cooling technique targets negative ions, which are typically resistant to cooling methods that work with atoms and positive ions. Read More »

Quantum Physics

### Synopsis: A Heat Engine Made of a Single Ion Spin

By converting electron spin into ion motion, researchers build a simple heat engine out of a single calcium ion. Read More »